Just received our first A5 for lab testing. Something is lose inside the radome and you can gear it rolling around when you move it. Has anyone else seen/heard this?
That is not typical. We will replace it for you. I’ll contact you privately for the details.
It seems to work fine, at least for now. Not sure if we need to RMA it but If you want it back for testing, let me know.